Recessed Patient Service Console shall be sized to accommodate all electrical and gas services specified.
Heavy gauge 6063-T5 extruded aluminum alloy frame complete with clear satin anodized finish and including an integral adjustable mounting plaster flange (1-7/8 inch adjustment). Topmounted die-formed ½ × ¾ inches conduit knockouts spaced 2.36 inches on center. Backplate and electrical service barriers to be constructed of minimum 16 gauge steel. Barriers to be provided to separate electrical services of different voltage or supply, and to separate medical gases from electrical services.
Minimum .08° 6063-T5 extruded aluminum alloy complete with clear satin anodized finish.